I have a patient who wears CIC hearing aids. He works in a loud environment and needs to hear when someone is talking to him. He currently wears earmuffs, but can't wear his hearing aids since they feed back. Is there any type of earmuff with a noise cancellation system that might help this patient? Any other suggestions?
Noise cancellation headsets are not the solution to this problem. More often called ANR (for active noise reduction), these devices can provide benefits in isolated instances with substantial low-frequency noise present, but are only occasionally of value in typical occupational settings and provide no obvious benefits for the hearing aid wearer. For more on the applicability of ANR in occupational noise see http://www.aearo.com/pdf/hearingcons/anr.pdf
Options for such a user are to wear an active sound-transmission earmuff which via use of a mic and internal speaker acts like a generic hearing aid at low sound levels, but gradually shuts down as the ambient noise increases, limiting transmitted sounds to the equivalent of 82-85 dBA. Another option is to turn the gain down on the aids and wear them under earmuffs with as large a cup size as possible to reduce the feedback problems. For the CIC user the E-A-R Ultra9000 earmuff also my be of value due to its external orifice that communicates directly to the concha via an acoustical duct. http://www.e-a-r.com/e-a-r.com/muff_detail.cfm?prod_family=Muff 9000&ind_prod_num=330-3028001 Additionally in select situations one may also consider wearing the aids as hearing protectors (see EARLog 18 at http://www.e-a-r.com/hearingconservation/earlog_main.cfm).
Conducting sound field tests with simulations of the actual ambient noise levels in which the patient typically works is also helpful to validate whatever approach is decided upon.
Elliott H. Berger Senior Scientist, Auditory Research E-A-R / Aearo Company 7911 Zionsville Road Indianapolis, IN 46268-1657 317-692-3031 w 317-692-3116 f
Elliott H. Berger, MS, INCE Bd. Cert.
After receiving his M.S. in Acoustical Engineering from North Carolina State University, Elliott joined the Aearo Company in 1976. As E•A•R/Aearo's Senior Scientist, Auditory Research, he conducts hearing protector research and development. He has written over 60 articles on hearing protection/conservation, and was the principal editor for the 4th and 5th editions of the AIHA Noise & Hearing Conservation Manual. Elliott chairs ANSI working group S12/WG11 on real-world hearing protector performance, and is also involved with numerous other standards committees. In 1993 he was the recipient of the National Hearing Conservation Association's Outstanding Hearing Conservationist Award, and in both 1993 and 1998 received their Outstanding Lecture Award. Elliott is Past President of the National Hearing Conservation Association (NHCA), Past-Chair of the American Industrial Hygiene Association's (AIHA) Committee on Noise and a Fellow of the Association, a Board Member of the of the Council for Accreditation in Occupational Hearing Conservation (CAOHC), a Fellow of the Acoustical Society of America (ASA), and a member of the American Speech-Language-Hearing Association (ASHA).
Amplification for patients in the music industry can be overwhelming and even a little confusing as these expert listeners hold different aural values that are often in opposition to their audiologists. This course will look at care for those patients in terms of needs, wants, and types of amplification, offering the audiologist information on culturally appropriate amplification.
Please note: You may earn ABA Tier 1 credits for this course if you complete it as part of the course 26346, "NIHL Webinar Series." Course 26346 contains recordings of all three events from our 2015 NIHL Webinar Series. ABA Tier 1 CEUs can be earned only when all modules are completed as part of course 26346.
'It gave me access to new information to present to my patients'Read Reviews
Join us for our eighth annual Noise-Induced Hearing Loss webinar series, organized by guest editor Brian Fligor, ScD, PASC and presented by leading experts. This modular series explores the latest research in noise-induced hearing loss and its considerations for audiology clinical practice. The courses take an indepth look at NIHL and its prevention and treatment when working with musicians and other audio professionals.
'very well thought outeasy to understanddid not talk over my head'Read Reviews
You are invited to see just how simple and informative the process of speech mapping can be when verifying the acoustical parameters of your hearing instrument fittings. Basic steps such as room and patient preparation, targets and measurements, and the application of data will be discussed with the work flow demonstrated using specific examples.
Psychoacoustic models for speech intelligibility and loudness can be used successfully for deriving a prescriptive fitting rule specifically to high frequency hearing losses. This article describes the development of a new prescriptive fitting algorithm for high frequency hearing losses, OPEN, including a detailed rationale and supporting clinical studies.
'I fit a lot of open fit hearing instruments and this course provided a lot of useful information'Read Reviews